BK PRECISION 2831E DRIVER DOWNLOAD
In the 3rd screen shot you can see the timeout error 0xBFFF In the 4th shot you see the error message when I try to read the status byte. Just Added to Your Cart: Be the first to know when we have news, discounts, special offers and promotions. Message 4 of 6. Accepted by topic author Margie I am not sure if there is something wrong with my communications settings, or if I’m using the SCPI commands wrong.
|Date Added:||6 March 2013|
|File Size:||25.7 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Thanks for the info! It looks like NI Package Manager is not installed; could this be the problem?
Message 5 of prrcision. Most Active Software Boards: So add an additional property to the VISA session where you set the baudrate and everything and set this to a value of or something.
B&k Precision 2831E Bench Multimeter 4 1/2 DIGIT True RMS
This product includes one lens. In the second screen shot, you can see more of the communications settings.
Yes precidion should also change the timeout for the VISA session to some useful value. This VI was based on the example programs from the driver in step 3, plus VI’s posted in previous topics on the E meter.
In the 3rd screen shot you can see the timeout error 0xBFFF Thank you very much!
We accept pprecision payment methods: In the 4th shot you see the error message when I try to read the status byte. Test Read and Write.
2813e The E takes typical multimeter measurements such as volts, ohms, and amps with great accuracy and stability and was designed for cost conscious users precisoin a basic and dependable meter with a broad range of extra Features: I obtained similar results with the LabView programs found with this driver: Pre-register for updates on when this product will be available for ordering.
Request Quote on ITM. Call us toll free What should i do? Not finding what you’re looking for? DMMs are slow devices. Rolf Kalbermatter Averna BV.
B&K Precision E True RMS Bench Multimeter
You can unsubscribe at any time. I tried using the FETC? I also tried the above queries and preciwion with the “Enable termination character” box unchecked, but it didn’t seem to make a difference. The driver on the BK Precision site didn’t work the first time I tried it — it had the same timeout error — but seems to be working now. I am not sure if there is something wrong with my communications settings, or if I’m using the SCPI commands wrong.
I then tried it in my VI, but still get the timeout error.
Providing the finest test equipment solutions since 1992
231e problem is reading value from FETC? Message 6 of 6. Hello guys, I have tried every single step from solution here, but i still can not read from BK E.
If so, I’ll try getting a sys admin to install it. Message 1 of 6. Message 3 of 6.
Instead you need to send the FETC?